Verification of test reference cords is easy, right? Yes, usually…but not always. After the power meter is set to 0 dB via the reference procedure, the loss of a mated connector pair is expected to be less than 0,20 dB for single-mode (e.g., -0,20 dB). Let’s examine one scenario where a loss is not negative (e.g., +0,15 dB).
The typical process for validating the loss of a connector, to verify if it is “reference grade” for example, is as follows: