Go ahead an install LinkWare PC 9,8 which can be obtained from here. While on that download page, download the Version 5,1 Build 4 firmware as well. Remember where you saved the firmware file!
- Make sure your battery is sufficiently charged; if in doubt, connect the power to your unit
- Open LinkWare PC 9,8
Connect the Versiv/TFS main frame to your PC using a Micro USB cord that came with your unit
Click on the Utilities > DSX CableAnalyzer > Software Update (You could of course click on OptiFiber Pro; for software updates - it makes no difference)
Locate the "update".cic file you downloaded and click on Open
This will send the update file to the unit, the update takes around 5 minutes.
You do not need to update the module - Versiv stores the firmware necessary to do any updates to the module automatically.
- If you have a remote unit (for example, used with DSX Copper modules or CFP modules), do the same for that too.