Information reported as summary result - CableIQ
|If you see Information as shown below: |
This is caused by the far end adapter not be connected or recognized.
You will also notice that CableID 4 has no length reported. This coax link was either terminated with a 75 ohm resistor or the length was in excess of 1.000 ft. Since the CableIQ does not have a limit for Length when testing coaxial cable, it will report Qualified.